The Miniature, Rugged Alternative to LVDTs and Linear Potentiometers
|Order • About • Site Map
Thermal Effect Calculator
What is the effect of a change in temperature on the accuracy of our position transducers? Is a temperature-compensation circuit required to account for the change in temperature in the operating environment?
For nearly all applications, the temperature effect on our position transducers is modest, if not negligible. The most significant temperature effect occurs on the stranded, Type 304 stainless steel displacement cable. The calculator above allows you to determine the effect of temperature change on the displacement cable length and position transducer's accuracy.
Incidentally, because our analog potentiometer-based products are designed to be used as voltage dividers (voltage variance) and not rheostats (resistance variance), there is no temperature effect related to the potentiometer itself. This is because there is no voltage change due to the temperature change: the potentiometer's resistance change occurs uniformly throughout the resistive material.
Any determination of thermal effect must also take into account the thermal effect on the object being measured.
The calculator uses this equation to calculate the change in length due to thermal expansion: delta l = a * l * delta T
The initial values above show a nominal 0.250-m position transducer range, a total of 0.375 m of displacement cable being used to attach to the application, 50° C of temperature change. Our standard displacement cables are constructed of Type 304 stainless steel and have a coefficient of thermal expansion of 0.00001728 m/m-°C (from 0°C to 100°C). This temperature change would increase the displacement cable length 0.000324 m in going from 0°C to 50°C. This results in an increase of ±0.0648% in inaccuracy over the full-scale measurement range of the position transducer.
Construction of Stainless Steel Wire Rope
No Warranties: This calculator and information are provided "as is" without any warranty, condition, or representation of any kind, either express or implied, including but not limited to, any warranty respecting non-infringement, and the implied warranties of conditions of merchantability and fitness for a particular purpose. In no event shall SpaceAge Control, Inc. be liable for any direct, indirect, special, incidental, consequential or other damages howsoever caused whether arising in contract, tort, or otherwise, arising out of or in connection with the use or performance of the information contained on this Web page.